Modbus RTU, Modbus TCP, Canbus, PLC's and Hmi's.

Call toll Free 1 887 877 4to 20mA or 1 877 486 2062

Modbus RTU, Modbus TCP, Canbus, PLC's and Hmi's.

Call toll Free 1 887 877 4to 20mA or 1 877 486 2062

The DAT5023 is Datexel's range of AC and DC current and voltage converters which are DIP switch programmable.

The DAT5023 I is an AC or DC current converter, the input current range is selectable on order from 0-5Amps, 0-10Amps, 0-20Amps, 0-25Amps, 0-30Amps, 0-40Amps, 0-50Amps, 0-60Amps. It measure the current with the use of a Hall effect transducer. There is a coil (current transformer) inside the current converter which is isolated from the current input. The cable passes through the housing of the converter and the coil and the true root mean square measurement is taken from this coil then converted to a standard industrial signal output.

The DAT5023 V has a AC and DC voltage input, it is selectable by connecting the voltage input to the correct terminals. Choose from 0-36 V, 0-80 V, 0-170 V, 0-370V and 0-550V. There is a DIP switch to select either AC or DC Voltage input.
The output is also selectable with DIP switches from 4-20mA, 0-10V, 0-20mA, 2-10V, 0-5V and 1-5V. The output current has an option of being passive or active output. The input, output and power supply are isolated from each other up to 2000 Vac. The power supply is between 18 and 30 Vdc and power consumption is less than 90mA.

DAT5023V AC

Input 0-36, 0-80, 0-170, 0-370, 0-550 Vac.

Output Current & Volts

DAT5023IAC A

Input AC Current 0-5 and 0-10 Amp

Output Current & Volts

DAT5023IAC B

Input AC Current 0-20, 0-25 Amp & 0-30 Amp

Output Current & Volts

DAT5023IAC C

Input AC Current 0-40, 0-50 & 0-60 Amp

Output Current & Volts

DAT5023V DC

Input 0-36, 0-80, 0-170, 0-370, 0-550 Vdc.

Output Current & Volts

DAT5023I DC A

Input DC Current 0-5 and 0-10 Amp

Output Current & Volts

DAT5023I DC B

Input DC Current 0-20, 0-25 Amp and 0-30 Amp

Output Current & Volts

DAT5023I DC C

Input DC Current 0-40, 0-50 and 0-60 Amp

Output Current & Volts

2000.35.014

Input AC/DC 0-25, 0-50 Amps

Output Modbus, 0-10V.

2000.35.013

Input AC/DC 0-25, 0-50 Amps

Output 4-20mA 2 wire