Modbus RTU, Modbus TCP, Canbus, PLC's and Hmi's.
Call toll Free 1 887 877 4to 20mA or 1 877 486 2062
The Intrinsically Safe Hockey Puck Temperature Transmitter DAT1015IS is an RTD Head Mount Temperature Transmitter. The input can accept mV's, Potentiometers and resistance Inputs. The DAT1015IS "in head Mount Temperature Transmitter" is PC Programmable. The sensor, range, zero and linearization, is programmable with the Prodat software, available from DATEXEL. The input is measured, linearized and converted to the industrial standard of 4-20mA signal. The RTD and resistance input has the added feature of selecting 3 or 4 wire for higher accuracy. There is a burn out failure alarm that can be programmed to drive the output signal High and Low. The input on the DAT1015IS has a high range of tolerance and the DAT1015IS can accept any voltage between 11 V and 30 Volt. The power is Loop Powered.
The DAT1015IS is accurate to 0.1% full scale mV Linear and has less than 0.02%/°f thermal drift.
The DAT1015IS is in compliance with the standard 89/336/CEE on the Electromagnetic Compatibility.
The DAT1015IS is housed in a self-extinguish plastic enclosure suitable for DIN B in-head mounting.
There is a mounting kit available to mount the DAT1015IS on DIN rail.
|Protection Mode:||II 1 G Ex ia IIC certified with Directive 94/9/CE ATEX|
|Input:||Thermocouples Type J, K, S, R, B, E, T and N.|
|RTD Pt100, Pt1000, Ni100, Ni1000, mV, Potentiometer,|
|Resistance. See Data sheet for range.|
|Power Supply:||10 to 32 Vdc|
|Temperature rating:||Operative- 4°F +158°F - 20°C +70°C|
|Temperature Storage Temperature - 40°F +185°F - 40°C +85°C.|
|Humidity:||(non condensing) 0-90%.|
|Dimensions:||Diameter 1 3/4 " Height 15/16 " 42 mm X 24mm|
|Weight:||3 oz 90 g.|
|EMC.||For industrial environments.|
|Immunity: EN 61000-6-2.|
|Emission: EN 61000-6-4.|
|Thermocouple||Positive 2 Negative 1|
|mV||Positive 2 Negative 1|
|RTD and Resistance 2 wires||Terminals 1 and 3|
|RTD and Resistance 3 wires||Terminal 1 and 3 Common 4|
|RTD and Resistance 4 wires||Terminal 1 Comp 2 and Terminal 3 Comp 4|
|Potentiometer||Terminal 1 and 3 Wiper 4|
|Output and Power supply||Positive +V Negative -V|